In this video you will learn about procedures on the care scape our 860. The suction procedure any ventilation mode may be used with a closed suction catheter the following alarms may occur during the use of. A closed suction procedure low expired minute ventilation low expired tidal volume and apnea to suction a patient without nuisance alarms an open suction procedure is provided by the ventilator select menu select suction.
And select start wait for the pre-oxygenation phase and then disconnect the patient at the. Y an audible tone sounds to indicate the patient is disconnected and ventilation is paused suction the patient reconnect the patient to resume ventilation the suction procedure will end after the oxygenation phase. Completes or if stop is selected to perform an original Iser. Treatment the ventilator supports the arrogant crow and arrogant solo inline nebulizers to connect the arrogant nebulizer to the Care scape our.
860 attach the nebulizer cable to the nebulizer line up the red dot from the nebulizer cable an arrogant port on. The vent and push in to connect the nebulizer time can be set for a specific duration of. Delivery times or for the volume of medication that will be delivered to the patient a general message shows the nebulizer treatment time remaining select menu select nebulizer.
Select arigon select time or continuous then select start to end the nebulizer treatment before set time select stop to perform a pneumatic nebulizer treatment the ventilator can compensate. For additional flow introduced by a pneumatic nebulizer into the patient’s circuit select menu select nebulizer select pneumatic select the flow value set the flow value to match the amount. Of flow that will be introduced into the circuit select start a general message shows when nebulizer flow compensation is on. Introduce the pneumatic nebulizer into the patient circuit for best results introduce the pneumatic nebulizer into the circuit within approximately 15 seconds after selecting start to end the nebulizer treatment.
Turn the pneumatic nebulizer flow source off then select stop to perform an increase Oh to procedure select menu select procedure select increase Oh to or press. The increase Oh – heart key use the factory default set increase Oh 2 percent or adjust the o2 percent to the desired concentration and confirmed the settings select start. The oxygen general message shows along with a progress bar and a two-minute countdown timer the procedure will end when.
The time has elapsed or when stop is selected or when the increase o – heart key is. Pressed you.